Catalog Number: 20737836322
For cobas c 311/501 analyzers:
In vitro test for the quantitative determination of digoxin in serum and plasma on Roche/Hitachi cobas c systems.
|Shelf life at 2‑8 °C:||See expiration date on cobas c pack label|
|On‑board in use and refrigerated on the analyzer:||26 weeks|
|Do not freeze.|
Specimen collection and preparation
Deselect Automatic Rerun for these applications in the Utility menu, Application screen, Range tab.
Calibration interval may be extended based on acceptable verification of calibration by the laboratory.
|Calibrators||S1‑6: Preciset TDM I calibrators|
|Calibration frequency||6‑point calibration• after lot change• as required following quality control procedures|
0.3‑5.0 ng/mL (0.38‑6.4 nmol/L) (defined by the Limit of Detection and the upper Limit of Linearity).
Representative performance data on the analyzers are given below. Results obtained in individual laboratories may differ.
Precision was determined using human samples and controls in a modified NCCLS EP5‑T2 protocol (repeatability n = 63, intermediate precision n = 63).The following results were obtained on a Roche/Hitachi cobas c 501 analyzer.
The following compounds were tested for cross‑reactivity.